Shanghai Haozhuang Instrument Co., Ltd. Haozhuang (LNB) brand intelligent light incubator is widely used in microbial tissue cell culture, seed germination, seedling experiment, plant culture, etc., can simulate different environmental climate conditions.
Common faults and maintenance of intelligent light incubator:
1. The shell of the incubator should be reliably grounded.
2. The incubator should be placed in a cool, dry, well-ventilated place away from heat sources and sunlight. Place it steadily to prevent vibration and noise.
3. In order to ensure the effective heat dissipation of the condenser, the distance between the condenser and the wall should be greater than 100mm. There should be a gap of 50mm on the side of the box and at least 300mm of space on the top of the box.
4. When carrying, repairing and maintaining the incubator, avoid collision, shaking and vibration; the maximum inclination is less than 45 degrees.
5. The instrument does not work suddenly, please check whether the fuse tube (behind the box) is burnt out, and check the power supply.
6. When cooling the incubator, it is not advisable to make the difference between the temperature inside the box and the ambient temperature greater than 25 degrees.
Common faults of intelligent light incubator
1. The compressor is not cooled, the temperature in the incubator is not uniform, or the circulation tuyere is frozen.
2. The compressor is always in working state under the balance mode, and the cooling effect is better and the compressor can be effectively maintained.
3. If it is a single-system operation, it will work only when the temperature drops to a certain value, and the temperature will fluctuate greatly when the door is opened and closed. Make the compressor ordinary
4. Starting is easy to shorten its service life.
5. Uneven temperature:
6. It is possible that the discharge items are too dense or the circulation fan in the box does not work
7. If the light is used for a long time, the light bulb will easily heat up and cause uneven temperature.
Icing at the air outlet:
Each refrigeration instrument will have a defrost cycle, during which the compressor will not cool. The heating element melts the ice on the condenser. The melted water is likely to gather in the air outlet and eventually cause ice.
Shanghai Haozhuang Instrument Co., Ltd. produces a series of light incubators. The high-quality and high-performance products are oriented to colleges and scientific research institutions across the country. The perfect service allows users to buy with ease and ease of use. Welcome new and old customers to inquire and purchase !
Door Bolt,High Quality Door Bolt,Door Bolt Details, Sunzhengde Hardware Mfg Co., Ltd.
Sunzhengde Hardware Mfg Co., Ltd. , https://www.sunzhengde.com